Benchtop temperature cycle test chamber is mainly applied for electrical, electronic products, components, and materials in the process of storage, transportation and use in high and low temperature environment, as well as temperature gradient test, but also for electronic components stress screening test, and the core component of the equipment is compressor, it will directly affect the performance of the equipment, many customers will consult how to maintain the equipment?
1. the voltage in the workplace of the high and low temperature test chamber must be stable. If the voltage is unstable, the current may be too large or too small. If the equipment works in this environment for a long time, it will bring great damage to the compressor. It is recommended to use a voltage stabilizer to connect the equipment.
2. the working environment of the high and low temperature test chamber should be well ventilated and dust free. In order to ensure that the compressor surface of the equipment is clean, the equipment should be placed in a dust-free space without use for a long time.
3. the high and low temperature test chamber in the process of use, do not start the compressor too frequently, and the compressor start time interval of at least 15 minutes.
Our company is proud to be able to design and manufacture climatic products for users. Through continuous improvement, progressive humanized management and effective after-sales service system, a unique business model has been gradually formed, as one of the enterprises with strength in the industry. From the research and development of new products to the effective use of new technology and the almost harsh pursuit of manufacturing technology to ensure the future reputation in the industry.